Tip induced fluorescence quenching for nanometer optical and topographical resolution
© Schulz et al.; licensee Springer. 2013
Received: 5 November 2012
Accepted: 24 January 2013
Published: 21 February 2013
Progress in nanosciences and life sciences is closely related to developments of high resolution imaging techniques. We introduce a technique which produces correlated topography and fluorescence lifetime images with nanometer resolution. Spot sizes below 5 nm are achieved by quenching of the fluorescence with silicon probes of an atomic force microscope which is combined and synchronized with a confocal fluorescence lifetime microscope. Moreover, we demonstrate the ability to locate and resolve the position of two fluorescent molecules separated by 20.7 nm on a DNA origami triangle with 120 nm side length by correlating topography and fluorescence data. With this method, we anticipate applications in nano- and life sciences, such as the determination of the structure of macromolecular assemblies on surfaces, molecular interactions, as well as the structure and function of nanomaterials.
KeywordsCorrelated atomic force and fluorescence microscopy Optical ultraresolution microscopy DNA nanostructures Single molecule detection
Nano- and life sciences progress in tandem with developments of high resolution imaging techniques. During the last decades, the invention and improvement of scanning probe microscopy techniques like atomic force microscopy (AFM) (Ando, 2012; Binnig et al., 1986; Engel & Muller, 2000; Garcia & Herruzo, 2012; Hinterdorfer et al., 2012) have offered new views onto nanoscale materials. Optical microscopy has been pushed beyond the diffraction limit and the observation of single molecules has evolved into a standard technology (Betzig et al., 2006; Dertinger et al., 2009; Dertinger et al., 2010b; Eggeling et al., 2009; Frey et al., 2006; Frey et al., 2004; Gustafsson, 2005; Hell, 2007; Hell & Wichmann, 1994; Hillenbrand & Keilmann, 2002; Moerner & Fromm, 2003; Rust et al., 2006; van de Linde et al., 2012). Several approaches for simultaneous high resolution topographical and optical imaging have been realized (Harke et al., 2012; Hartschuh, 2008; Novotny, 2007; Noy & Huser, 2003; Yang et al., 2000). Most of these high resolution approaches exploit local field enhancement effects near sharp tips to increase the fluorescence emission of fluorophores in close proximity (Hartschuh, 2008). Optical resolutions below 10 nm are difficult to achieve. In a rare example, Quake et al. reported an optical resolution of 8 nm using tip induced fluorescence enhancement (Ma et al., 2006). However, enhancement of fluorescence depends strongly on the orientation of the molecule to the probe, which introduces uncertainty in the location and respective distance of molecules of different orientation. Xie and Dunn, and Ambrose et al. combined near field scanning optical microscopy (NSOM) and fluorescence lifetime imaging microscopy (FLIM) using metal coated near field probes to study the fluorescence emission of single dye molecules on surfaces with spatial resolutions in the range of about 100 – 300 nm (Ambrose et al., 1994; Xie & Dunn, 1994). They observed a dependence of the fluorescence lifetime from the relative position of the near field probe, indicating quenching due to the metal coating.
In this work, we take advantage of localized fluorescence quenching of sharp silicon tips to create a novel high resolution imaging technique. We exploit this effect to spatially and temporally correlate optical and topographical imaging at sub 5 nm resolution. Standard, non-modified AFM tips can be used for this tip induced quenching imaging (TIQI) ensuring high optical, as well as topographical resolutions. We present measurements with individual fluorophores and DNA origami (Rothemund, 2006) triangles with 120 nm sides into which two fluorophores are incorporated at 20.7 nm from each other. We were able to resolve the morphology of the DNA triangle and to confirm the localization and distance of the fluorophores.
The energy of an electronically excited fluorophore can either be used to emit a photon (radiation), or the energy can be transferred to another object or the environment (i.e. fluorophore, surface, solvent molecule, etc.) resulting in non-radiative decay. Several mechanisms are known to alter the fluorescence of a molecule by changing both the radiative and the non-radiative pathways. Quenching is the term used whenever this alteration results in detection of less fluorescence light. The process behind quenching can amongst others be explained by either dipole-dipole interaction, direct energy or electron transfer (Dexter, 1953; Forster, 1948). It has been shown that quenching can be utilized to probe distances in the nanometer or sub-nanometer range (Berndt et al., 2010; Marme et al., 2003; Selvin, 2000). The impact of metal tips or nanostructures close to fluorophores has been well characterized (Ambrose et al., 1994; Anger et al., 2006; Ebenstein et al., 2006; Eckel et al., 2007; Krug et al., 2005; Kuehn et al., 2006; Xie & Dunn, 1994; Yoskovitz et al., 2011). Here, we exploit the quenching of organic fluorophores by sharp n-doped silicon tips to record topographical and optical images simultaneously with nanometer resolution.
The experimental setup consists of a MicroTime200 confocal fluorescence lifetime microscope (PicoQuant) with an inverted optical microscope (Olympus IX-71) onto which a sample scanning atomic force microscope (MFP-3D, Asylum Research) is mounted (Koenig et al., 2009). The setup, synchronization, and alignment have been described in detail (Schulz et al., 2010). Pulsed diode lasers were used for excitation at 640 nm and 470 nm. The polarizations were linear along the slow scan axis (vertical in the images) for the single fluorophore images and circular for the DNA origamis. The circular polarization increased the chance to excite both fluorophores on one DNA origami equally. The alignment of the AFM probe with the confocal volume is realized by monitoring the laser light that is reflected back from the sample surface. Imaging the back reflected light creates a specific pattern, which is distorted by the AFM tip. For alignment, the objective is moved until the distortion is centered in this pattern. This procedure is quick, requires only low laser intensity (to reduce photo bleaching) and is reliably accurate within 10 nm.
For the combined AFM/FLIM experiments with DNA origami triangles, a thin sheet of muscovite mica was deposited on a glass coverslide as described (Ma et al., 2006). These surfaces were incubated with 10 μl of a solution of TAE buffer containing 10 μM Mg2+ and 0.4 nM DNA origami for 5 min and then successively rinsed with 200 μl TAE buffer, 200 μl MgCl2 (200 mM) in water, and 400 μl ethanol and then dried in a stream of nitrogen gas. The origami samples were prepared as described in (Rothemund, 2006) and Additional file 1. For the single dye experiments, a drop of solution containing the fluorophores was deposited on a glass cover slide to give a density of active fluorophores of about 1 molecule per 1 μm2. The drop was allowed to dry on the surface, and the sample was used without further treatment. All combined experiments were performed under ambient conditions in contact mode.
Results and discussions
High resolution scans reveal a quenching efficiency of up to 100% within background noise. Not only a change in the fluorescence intensity but also a change in fluorescence lifetime can be observed (Figure 2f). This result provides some physical insights into the effects that dominate the interaction between the fluorophore and the AFM tip. In the outer region of the dark spot, we find an area of shortened fluorescence lifetime which indicates that non-radiative energy transfer plays a role at very short distances from the tip (Aussenegg et al., 1987). The intensities and lifetimes at distances further from the tip are consistent with a radiative dipole - induced dipole type feedback mechanism (Barnes, 1998) and geometrical considerations as outlined in Figure 2g. The fluorescence of a molecule near a surface depends strongly on the orientation of its dipole relative to the interface. The method of image dipoles can be used graphically to exploit the impact of this effect. From defocused imaging, (Marshall et al., 2011) we could show that the molecular transition dipole moments lie parallel to the sample surface (see Additional file 1). When the fluorophore is directly under the silicon tip and thus parallel to the tip surface, it creates an antiparallel oriented mirror dipole inside the tip. This creates an electric field that leads to negative feedback. In the case that the fluorophore is beside the tip, its dipole has a component normal to the silicon surface leading to positive feedback (Figure 2g). For very short distances also the vertical case results in quenching because of direct energy transfer to the silicon. The high localization of the quenched spot and the asymmetry in the data is related to the orientation between the tip and fluorophore, which changes swiftly as the molecule is scanned relative to the AFM tip (Figure 2g). We can evaluate this model by solving Maxwell’s equations in planar structures in the presence of an oscillating electric dipole source (the emitting molecule) by using Sommerfeld integrals (Chizhik et al., 2009; Enderlein, 2000; Enderlein & Ruckstuhl, 2005) and taking into account the change of the relative orientations of dipole and AFM tip surface during the scanning motion (Additional file 1). Comparison of our experimental data and the calculated fluorescence intensity images obtained using this method show good agreement (Figure 2f and Additional file 1: Figure S4).
The interaction between the AFM probe and the fluorophore would be elucidated further by investigating the fluorescence response with changing tip-surface distance. In our experiments under ambient conditions, however, it was impossible to gather such data: strong capillary forces between the AFM probe and the surface dominate for short distances. Thus, when one approaches the AFM tip to the surface, a “snap-in” event can be observed; that means, at some distance, capillary forces start to dominate, and the AFM tip moves very quickly to the surface, within a time that cannot be resolved in the fluorescence data. When retracting the AFM tip from the surface, the opposite effect occurs, and the AFM probe is held on the surface until the restoring force of the cantilever exceeds the capillary forces. The length scale, at which the snap-in and snap-back occur, coincide with the length scale where the fluorescence is strongly modulated, such that no interesting information could be extracted from the force/fluorescence-distance curves that we acquired.
The quenching mechanism does not depend on the use of particular organic fluorophores. We observed fluorescence quenching with silicon tips for several different classes of dyes over a range of excitation and emission wavelengths. We tested successfully the oxazine derivative Atto655, as well as the cyanine Alexa647, and the rhodamine derivatives Atto647N and Alexa488 (Additional file 1: Figure S5).
Interestingly, for quantum dots embedded in 10 nm thin PMMA films, Ebenstein et al. did not observe quenching for bare silicon tips (Ebenstein et al., 2006). This may be due to the larger distance between the tip and the quantum dot and/or the thickness of the quantum dot shell.
The fluorescence intensity image in Figure 3b shows the overlay of the undistinguishable diffraction limited spots of the two dye molecules. A pattern that shows the outer- and innermost DNA strands in the structure and the position of the fluorophores on the DNA is shown superimposed on the AFM image in Figure 3d. This pattern is also shown superimposed on the fluorescence intensity image to compare the position of the quenched spots with the expected position determined from the topography image. Figure 3e shows that the quenched spots, within the experimental uncertainty, indeed coincide in position with the location of the fluorophores according to the topography image. The green circles in Figure 3 represent a 10 nm uncertainty based on the limitation of our optical and topographical resolution. Because of the overlapping diffraction limited spots, the fluorescence intensity in the dips falls to about 50% of the signal. It is worth noting that the topography image and the quenching image were not translated to match each other. This is possible because the two microscopes are synchronized, and the data is sorted into pixels according to the time at which it was acquired (Schulz et al., 2010). Since the tip records the topography at the molecule’s position and changes its fluorescence at the same time, these features appear in the same position in both images.
Mica surfaces have a rather strong influence on the emission of the fluorophores; blinking and bleaching are much more prominent compared to the single dye molecules on glass. In fact, many of the DNA origamis that can be seen in the AFM images do not show fluorescence above background level. Also, even when both fluorophores of an origami are emitting during the scan, if one or both of them blink while the AFM probe is near the fluorophore, the quenching would be obscured. Another problem arises when DNA origamis with their fluorophores are too close to each other, which reduces the signal to noise ratio. Considering eight triangles, for which the two quenched spots could be clearly distinguished, we found an average distance between the fluorophores of 21 nm ± 2 nm (s.d.) by measuring the distances between the two minima of the fluorescence intensities in sections like Figure 3f. This matches exceptionally well the expected value of 20.7 nm from the DNA design.
In this work, we have presented a novel method for simultaneous optical and topographical resolutions in the sub 5 nm range. This range bridges the atomic and the optical superresolution range, making it possible to address important questions in supramolecular chemistry, molecular biology, and nanosciences. The combination of topography and fluorescence adds high resolution chemical contrast to the high topographical resolution of AFM. This approach could be extended to multicolor imaging and to imaging under liquid conditions. The commercial availability of the instruments and the AFM tips should allow this method to spread rapidly into other fields.
Atomic force microscopy
Fluorescence lifetime imaging microscopy
Tip induced quenching imaging
Full width at half maximum.
This work was supported by Arizona State University, ONR (N000140911118) and ASU Presidential Strategic Initiative Fund. The authors thank Dr. Anne K. Jones for proofreading the manuscript.
- Ambrose WP, Goodwin PM, Martin JC, Keller RA: Alterations of single-molecule fluorescence lifetimes in near-field optical microscopy. Science 1994, 265: 364–367. 10.1126/science.265.5170.364PubMedView Article
- Ando T: High-speed atomic force microscopy coming of age. Nanotechnology 2012, 23: 062001. 10.1088/0957-4484/23/6/062001PubMedView Article
- Anger P, Bharadwaj P, Novotny L: Enhancement and quenching of single-molecule fluorescence. PhysRevLett 2006, 96: 113002.
- Aussenegg FR, Leitner A, Lippitsch ME, Reinisch H, Riegler M: Novel aspects of fluorescence lifetime for molecules positioned close to metal-surfaces. Surf Sci 1987, 189: 935–945.View Article
- Barnes WL: Fluorescence near interfaces: the role of photonic mode density. J Mod Opt 1998, 45: 661–699. 10.1080/09500349808230614View Article
- Berndt M, Lorenz M, Enderlein J, Diez S: Axial nanometer distances measured by fluorescence lifetime imaging microscopy. Nano Lett 2010, 10: 1497–1500. 10.1021/nl100593xPubMedView Article
- Betzig E, Patterson GH, Sougrat R, Lindwasser OW, Olenych S, Bonifacino JS, Davidson MW, Lippincott-Schwartz J, Hess HF: Imaging intracellular fluorescent proteins at nanometer resolution. Science 2006, 313: 1642–1645. 10.1126/science.1127344PubMedView Article
- Binnig G, Quate CF, Gerber C: Atomic force microscope. PhysRevLett 1986, 56: 930–933.
- Chizhik A, Schleifenbaum F, Gutbrod R, Khoptyar D, Meixner AJ, Enderlein J: Tuning the fluorescence emission spectra of a single molecule with a variable optical subwavelength metal microcavity. Phys Rev Lett 2009, 102: 073002–073006.PubMedView Article
- Dertinger T, Colyer R, Iyer G, Weiss S, Enderlein J: Fast, background-free, 3D super-resolution optical fluctuation imaging (SOFI). P Natl Acad Sci USA 2009, 106: 22287–22292. 10.1073/pnas.0907866106View Article
- Dertinger T, Colyer R, Vogel R, Enderlein J, Weiss S: Achieving increased resolution and more pixels with superresolution optical fluctuation imaging (SOFI). Opt Express 2010, 18: 18875–18885. 10.1364/OE.18.018875PubMed CentralPubMedView Article
- Dertinger T, Heilemann M, Vogel R, Sauer M, Weiss S: Superresolution optical fluctuation imaging with organic dyes. Angew Chem Int Ed Engl 2010, 49: 9441–9443. 10.1002/anie.201004138PubMed CentralPubMedView Article
- Dexter DL: A theory of sensitized luminescence in solids. J Chem Phys 1953, 21: 836–850. 10.1063/1.1699044View Article
- Ebenstein Y, Yoskovitz E, Costi R, Aharoni A, Banin U: Interaction of scanning probes with semiconductor nanocrystals; physical mechanism and basis for near-field optical imaging. J Phys Chem A 2006, 110: 8297–8303. 10.1021/jp056229oPubMedView Article
- Eckel R, Walhorn V, Pelargus C, Martini J, Enderlein J, Nann T, Anselmetti D, Ros R: Fluorescence-emission control of single CdSe nanocrystals using gold-modified AFM tips. Small 2007, 3: 44–49. 10.1002/smll.200600130PubMedView Article
- Eggeling C, Ringemann C, Medda R, Schwarzmann G, Sandhoff K, Polyakova S, Belov VN, Hein B, von Middendorff C, Schoenle A, Hell SW: Direct observation of the nanoscale dynamics of membrane lipids in a living cell. Nature 2009, 457: 1159–1163. 10.1038/nature07596PubMedView Article
- Enderlein J: A theoretical investigation of single-molecule fluorescence detection on thin metallic layers. Biophys J 2000, 78: 2151–2158. 10.1016/S0006-3495(00)76761-0PubMed CentralPubMedView Article
- Enderlein J, Ruckstuhl T: The efficiency of surface-plasmon coupled emission for sensitive fluorescence detection. Opt Express 2005, 13: 8855–8865. 10.1364/OPEX.13.008855PubMedView Article
- Engel A, Muller DJ: Observing single biomolecules at work with the atomic force microscope. NatStructBiol 2000, 7: 715–718.
- Forster T: Zwischenmolekulare Energiewanderung und Fluoreszenz. Ann Phys-Berlin 1948, 2: 55–75.View Article
- Frey HG, Witt S, Felderer K, Guckenberger R: High-resolution imaging of single fluorescent molecules with the optical near-field of a metal tip. Phys Rev Lett 2004, 93: 200801.PubMedView Article
- Frey HG, Bolwien C, Brandenburg A, Ros R, Anselmetti D: Optimized apertureless optical near-field probes with 15 nm optical resolution. Nanotechnology 2006, 17: 3105–3110. 10.1088/0957-4484/17/13/004View Article
- Garcia R, Herruzo ET: The emergence of multifrequency force microscopy. Nat Nanotechnol 2012, 7: 217–226. 10.1038/nnano.2012.38PubMedView Article
- Gerton JM, Wade LA, Lessard GA, Ma Z, Quake SR: Tip-enhanced fluorescence microscopy at 10 nanometer resolution. Phys Rev Lett 2004, 93: 180801.PubMedView Article
- Gustafsson MGL: Nonlinear structured-illumination microscopy: wide-field fluorescence imaging with theoretically unlimited resolution. P Natl Acad Sci USA 2005, 102: 13081–13086. 10.1073/pnas.0406877102View Article
- Harke B, Chacko JV, Haschke H, Canale C, Diaspro A: A novel nanoscopic tool by combining AFM with STED microscopy. Optical Nanoscopy 2012, 1: 3. 10.1186/2192-2853-1-3View Article
- Hartschuh A: Tip-enhanced near-field optical microscopy. Angew Chem Int Ed Engl 2008, 47: 8178–8191. 10.1002/anie.200801605PubMedView Article
- Hell SW: Far-field optical nanoscopy. Science 2007, 316: 1153–1158. 10.1126/science.1137395PubMedView Article
- Hell SW, Wichmann J: Breaking the diffraction resolution limit by stimulated-emission-depletion fluorescence microscopy. Opt Lett 1994, 19: 780–782. 10.1364/OL.19.000780PubMedView Article
- Hillenbrand R, Keilmann F: Material-specific mapping of metal/semiconductor/dielectric nanosystems at 10 nm resolution by backscattering near-field optical microscopy. Appl Phys Lett 2002, 80: 25–27. 10.1063/1.1428767View Article
- Hinterdorfer P, Garcia-Parajo MF, Dufrene YF: Single-molecule imaging of cell surfaces using near-field nanoscopy. Acc Chem Res 2012, 45: 327–336. 10.1021/ar2001167PubMedView Article
- Huang B, Bates M, Zhuang XW: Super-resolution fluorescence microscopy. Annu Rev Biochem 2009, 78: 993–1016. 10.1146/annurev.biochem.77.061906.092014PubMed CentralPubMedView Article
- Koenig M, Koberling F, Walters D, Viani J, Ros R, Schulz O: Combining atomic force microscopy with confocal microscopy. 2009. Accessed 19 Dec 2012 http://www.picoquant.com/technotes/technote_mt200_afm.pdf;
- Krug JT, Sanchez EJ, Xie XS: Fluorescence quenching in tip-enhanced nonlinear optical microscopy. Appl Phys Lett 2005, 86: 233102. 10.1063/1.1935769View Article
- Kuehn S, Hakanson U, Rogobete L, Sandoghdar V: Enhancement of single-molecule fluorescence using a gold nanoparticle as an optical nanoantenna. Phys Rev Lett 2006, 97: 017402.View Article
- Ma ZY, Gerton JM, Wade LA, Quake SR: Fluorescence near-field microscopy of DNA at sub-10 nm resolution. Phys Rev Lett 2006, 97: 260801.PubMedView Article
- Marme N, Knemeyer JP, Sauer M, Wolfrum J: Inter- and intramolecular fluorescence quenching of organic dyes by tryptophan. Bioconjugate Chem 2003, 14: 1133–1139. 10.1021/bc0341324View Article
- Marshall GD, Gaebel T, Matthews JCF, Enderlein J, O’Brien JL, Rabeau JR: Coherence properties of a single dipole emitter in diamond. New J Phys 2011, 13: 055016. 10.1088/1367-2630/13/5/055016View Article
- Moerner WE, Fromm DP: Methods of single-molecule fluorescence spectroscopy and microscopy. Rev Sci Instrum 2003, 74: 3597–3619. 10.1063/1.1589587View Article
- Novotny L: Chapter 5 the history of near-field optics. In Progress in optics Vol. Volume 50 edition. Edited by: Wolf E. 2007, 137–184. ElsevierView Article
- Noy A, Huser TR: Combined force and photonic probe microscope with single, molecule sensitivity. Rev Sci Instrum 2003, 74: 1217–1221. 10.1063/1.1536260View Article
- Rothemund PWK: Folding DNA to create nanoscale shapes and patterns. Nature 2006, 440: 297–302. 10.1038/nature04586PubMedView Article
- Rust MJ, Bates M, Zhuang X: Sub-diffraction-limit imaging by stochastic optical reconstruction microscopy (STORM). Nat Meth 2006, 3: 793–795. 10.1038/nmeth929View Article
- Schulz O, Koberling F, Walters D, Koenig M, Viani J, Ros R: Simultaneous single molecule atomic force and fluorescence lifetime imaging. Proc SPIE 2010, 7571: 757109. 10.1117/12.842502View Article
- Selvin PR: The renaissance of fluorescence resonance energy transfer. NatStructBiol 2000, 7: 730–734.
- Steinhauer C, Jungmann R, Sobey TL, Simmel FC, Tinnefeld P: DNA origami as a nanoscopic ruler for super-resolution microscopy. Angew Chem Int Ed Engl 2009, 48: 8870–8873. 10.1002/anie.200903308PubMedView Article
- van de Linde S, Heilemann M, Sauer M: Live-cell super-resolution imaging with synthetic fluorophores. Annu Rev Phys Chem 2012,63(63):519–540.PubMedView Article
- Westphal V, Kastrup L, Hell SW: Lateral resolution of 28 nm (lambda/25) in far-field fluorescence microscopy. ApplPhysB 2003, 77: 377–380.
- Xie XS, Dunn RC: Probing single-molecule dynamics. Science 1994, 265: 361–364. 10.1126/science.265.5170.361PubMedView Article
- Yang TJ, Lessard GA, Quake SR: An apertureless near-field microscope for fluorescence imaging. Appl Phys Lett 2000, 76: 378–380. 10.1063/1.125759View Article
- Yoskovitz E, Menagen G, Sitt A, Lachman E, Banin U: Nanoscale near-field imaging of excitons in single heterostructured nanorods. Nano Lett 2010, 10: 3068–3072. 10.1021/nl101614sPubMedView Article
- Yoskovitz E, Hadar I, Sitt A, Lieberman I, Banin U: Interplay of quenching and enhancement effects in apertureless near-field fluorescence imaging of single nanoparticles. J Phys Chem C 2011, 115: 15834–15844. 10.1021/jp2035604View Article
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.